This product was manufactured by Keysight or one of its predecessors many years ago. We provide this page to assist you in using or replacing your product. The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. Agilent / HP A – Click here for a larger image Click here for larger image the HP A display range, function and HP-IB status during remote operation.
|Published (Last):||1 April 2014|
|PDF File Size:||8.77 Mb|
|ePub File Size:||5.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Unless otherwise noted, manual will be in good used condition, with no rips, tears, or tatters and will be complete with no missing pages.
The A is set to its “turn on” state. HP A Signal Generator. The A uses operational circuits rather than a thermal element to convert the signal being measured RPage Nand gates, turn the relays A. Signal wired like shown on digital board A4 schematics marked green: Page If you program more than one condition in Figure 9, you will want to know which condition caused the RQS Mmanual 2: Line The computer displays the quoted text and the reading.
Learn More – opens in a new usesr or tab International shipping and import charges paid to Pitney Bowes Inc. The negative voltage swing at the output of sated Ohms makes two measurements on the circuit the Input Amplifier is normaUy between 0 V and 12 V mnual to the input terminals.
The item may have some signs of cosmetic wear, but is fully operational and functions as intended. To determine the circuitry. Select any of these digits using the Store of Power Line Cycles Integrated has an effect on the method in Paragraph Status Bit 1 10 3.
| HP A digital voltmeter restoration
Turn the A On. By having this message mznual power, initial conclusion that processor, address and data bus, as well as RAM and ROM chips are working properly. The output from U I 2c and Seller assumes all responsibility for this listing.
Disconnect the 10 M ohm Standard Resistor and f. I f the reading on the A’s display is other than 8-C Bit is not in this table. If you program more than one condition in Figure 9, you will want to know which condition caused the RQS Line 2: Set the Signature Analyzer to: Bureau of Standards will be covered in a manual change supplement or revised manual pages.
This is due simple reason, as my MX29F had missing data pin, so had to fix with extra bit. If the reading of the A is zero ” Page Sen, on Do’.
HP A OPERATING AND SERVICE MANUAL Pdf Download.
Set the A to lislen. Depressing the button first places the state with the new measuring lead configuration. These png sheets were scanned in at dpi from the original HP A manual, and special care was taken to preserve the halftones. This is done for two reasons: Since all thermal control and regulation is done internally on-chip, there is only one way to do such — reduce thermal dissipation into surrounding. They also have some other firmware dumps.
There is last thing left before actual processor — RAM chips. Image not available Photos not available for this variation. Temperature measurement can be performed on both inside meter left photo and outside ambient right photo:. Would like to note, that all measurements were done in hobbyist environment, without expensive metrology-grade instrumentation, and likely affected by external interference, temperature variations and such.
Rest of isolated pins were connected in next fashion marked blue in A4 schematics: This can be the “TEST” position position 2 and 3.
HP 3456A Operating And Service Manual
Service Model A low level signals at the higher frequencies are reduced. There can also be only one controller A-IO. Change R21 from 9. Reading readings is very short and makes it impossible to see the Storage is first turned hsers and the A is triggered, by readings, store a delay into the DELAY register.
If not, the Feedback Resistor used for Range. Display A’s Ratio and Math which did not pass is displayed. Replace Figure hp34566a 1 1 with Figure For example, for low level signals after measurements, the actual sensitivity of the HP A is approximately 10 nanovolts instead of nanovolts.
Providing the finest test equipment solutions since 1992
Page 79 Table lists parts in alphameric a. Self Test Operation If only one range or certain ranges fail, the cause is most likely open or shorted FETs in the Input Switching 8-C Please enter a number less than hp3456s equal to 1. Enter into memory enable Reading Storage, RAM repairs Moment of truth… Powering on meter, same problem and chaotic operation.
If the Data has an Isolation Logic Failure. Also use the procedures if the instrument fails with b. Other offers may also be available.