Find the most up-to-date version of ASTM C at Engineering ASTM. C Standard Test Method for Dynamic Young’s Modulus, Shear Modulus, and Poisson’s Ratio for Advanced Ceramics by Impulse Excitation of. ASTM C Advanced ceramics, impact excitation. EN, EN Advanced technical ceramics, ultrasonics, resonance and impact.
|Published (Last):||19 February 2004|
|PDF File Size:||1.36 Mb|
|ePub File Size:||16.78 Mb|
|Price:||Free* [*Free Regsitration Required]|
Are you sure you want to empty the cart? The rod and bar geometries are described in the main body. Flat plates and astj may also be measured similarly, but the required equations for determining the moduli are not addressed herein. The specimens are subjected to minute strains; hence, the moduli are measured at or near the origin of the stress-strain curve, with the minimum possibility of fracture. If you need to be able to add standards to the monitoring service, you must register as a user with your own username and password.
A transducer for example, contact accelerometer or non-contacting microphone senses the resulting mechanical vibrations of the specimen and transforms them into electric signals.
Please first verify your email before subscribing to alerts. Read more about subscriptions.
The following editions for this book are also available Standards monitoring is a free service designed for you who wish to be alerted of changes to or new editions of the standards required in your work. Dynamic Young’s modulus is determined using the resonant frequency in the flexural mode of vibration.
Dynamic Young’s modulus and dynamic shear modulus are used to compute Poisson’s ratio.
This standard is also available to be included in Standards Subscriptions. C Standard Classification of Mullite Refractories. C12259 to Alert PDF. Any specimen with a resonant frequency falling outside the prescribed frequency range is rejected.
Need more than one copy?
Your Alert Profile lists the documents that will be monitored. If the document is revised or amended, you will be notified by email. Search book title Enter keywords for book title search. NOK 1 ,00 excl. Dynamic Young’s modulus is determined using the resonant frequency in the flexural mode of vibration. The dynamic elastic properties of a material can therefore be computed if the geometry, mass, and mechanical resonant frequencies of a suitable rectangular, cylindrical, or disc geometry test specimen of that material can be measured.
The moment new editions of the standards you monitor are released, changes are made, or appendixes are published, you will be alerted by email. The disc geometry is addressed in Annex A1. No items in cart. Click here to download full list of books.
Specimens of these materials possess specific mechanical resonant frequencies that are determined by the elastic modulus, mass, and geometry of the test specimen. The disc geometry is addressed in. Deviations in specimen dimensions or mass and internal astmm cracks, delaminations, inhomogeneities, porosity, etc will change the resonant frequency for that specimen.
The rod and bar geometries are described in the main body. There were no books found for the applied search filters. Standards Subscriptions from ANSI provides a money-saving, multi-user solution for accessing standards. FORMAT Single User Subscription Site License Subscription BOOKS All Books PACKAGES All Packages EDITION to PUBLISHER.
Enter your account email address to request a password reset: Webprint Printed and bound. The rod and bar geometries are described in the main body.
It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
A transducer for example, contact asfm or non-contacting microphone senses the resulting mechanical vibrations of the specimen and transforms them into electric signals. The dynamic shear modulus, or modulus of rigidity, is found using torsional resonant vibrations.