View and Download HP A manual change online. Impedance/Gain-Phase. A Multimeter pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Manual Change. Current Directory: /HP Agilent/HP A Impedance Gain Phase Analyzer. Click on a File to download or a Directory to open. File Type. File Size. ===> Back.
|Published (Last):||27 September 2018|
|PDF File Size:||15.82 Mb|
|ePub File Size:||8.62 Mb|
|Price:||Free* [*Free Regsitration Required]|
The problem is that manhal have to manipulate the circuit element with an external AC field via a function generator and simultaneously measure the circuit properties of the active nanostructure as a function of frequency using the impedance analyzer. Can you use a DC 2. Message 11 of PV charger battery circuit 4.
If I actuated from say Hz – 1 kHz using the analyzer, it would change the properties of the device. Message 15 of AF modulator in Transmitter what is the A?
Service manual for an HP A Impedance/Gain-Phase Analyzer
When the electric field is turned off, the “wires” break apart and does not contact the electrodes anymore. Message 13 of View Cart My Account. Using HP A’s Equivalent Circuit Analysis Function, you can easily and quickly obtain those equivalent circuit constants that, until now, required a number of time-consuming, complicated calculations. The HP A’s ASP function, an internal programming feature, allows you 41194a control all HP A operations measurement, display, and analysis without the need for an external computer.
HP 4194A Manual Change
I think that the analyzer is essentially a function generator. Dec 248: Calibration accuracy is guaranteed to the tips of the HP A 1.
This makes it possible to make highly accurate measurements basic measurement accuracy 1. High accuracy and wide range- Impedance measurement: You could use the HP only, set the bias voltage to 2. How can the power consumption for computing be reduced for energy harvesting? Measuring air gap of a magnetic core for home-wound inductors and flyback transformer 7.
HP ServMan http: How do you get an MCU design to market quickly?
HP A MANUAL CHANGE Pdf Download.
I have a better idea now, but I am still not real clear on what you are doing. Maybe you can explain why you chose k to k. The device acts as a resistor for frequencies greater than kHz and when the AC field is turned off it loses its resistive behavior. The device I am testing is a planar gold electrode system. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
4194A Impedance/Gain-Phase Analyzer
The frequency sweep done by the HP analyzer does not overlap the AC frequency applied by the function generator. I am not sure that there is a way where you can separate the interference of the sine function from the Imp analyzer and the constant 2MHz from your FG. Losses in inductor of a boost converter 9. Hierarchical block is unconnected 3.
Heat sinks, Part 2: Without getting into the particular details of the experiment, an AC field is applied to the electrodes using a function generator, to form “wires” between the electrodes bridging the gap.
The HP A’s high degree of measurement accuracy – 0. The HP A makes high-speed measurements, approximately 3. Detailed impedance and transmission characteristics, including secondary parameter derivations, can be simply and quickly evaluated or tested.
Hey Brian Thanks a lot for the quick reply. If its circuit really good readable is?